[Histonet] Sakura Tissue-tek Glas Air bubble problems
flnails <@t> texaschildrens.org
Tue May 17 09:33:40 CDT 2011
If you are using one of the older film coverslippers there is a roller following the application of the film.
Make sure that roller is still in the correct place or that one of the springs is not missing or in need of replacing.
The job of that roller is to squeeze the air bubbles from under the film.
From: histonet-bounces <@t> lists.utsouthwestern.edu [mailto:histonet-bounces <@t> lists.utsouthwestern.edu] On Behalf Of Lees, Sara
Sent: Tuesday, May 17, 2011 9:18 AM
To: 'histonet <@t> lists.utsouthwestern.edu'
Subject: [Histonet] Sakura Tissue-tek Glas Air bubble problems
I am having horrible problems with air bubbles , can anyone help I have changed the speed increased the mountant volume, clean it, kissed it, u name it and nothing has worked...Can anyone help
Confidentiality Notice: This e-mail transmission may contain confidential or legally privileged information that is intended only for the individual or entity named in the e-mail address. If you are not the intended recipient, you are hereby notified that any disclosure, copying, distribution, or reliance upon the contents of this e-mail is strictly prohibited.
If you have received this e-mail transmission in error, please reply to the sender, so that we can arrange for proper delivery, and then please delete the message from your inbox. Thank you.
The information in this e-mail may be confidential and/or
privileged. If you are not the intended recipient or an
authorized representative of the intended recipient, you
are hereby notified that any review, dissemination, or
copying of this e-mail and its attachments, if any, or
the information contained herein is prohibited. If you
have received this e-mail in error, please immediately
notify the sender by return e-mail and delete this e-mail
from your computer system. Thank you.
More information about the Histonet