[Histonet] Leica Service Technician&In-Reply-To=
ruppert.amysue <@t> marshfieldclinic.org
Wed Aug 3 17:10:37 CDT 2011
We moved into a new lab a few years ago, and due to the incoming air current with the ventilation here, it caused our cyrostats to ice up considerably. We have the type of ventilation that has air vents on the ceiling and are constantly blowing large amounts of air into our rooms. We had to get a piece of plastic put into the vent to divert some of the air away from our cryostat. It helped, but still ices more than when we were in our old lab. Anyway, point is to check out the air flow above the cyrostat. You also need to have enough clearance behind the cryostat and on each side. The clearance amount would be listed in the manual.
The contents of this message may contain private, protected and/or privileged information. If you received this message in error, you should destroy the e-mail message and any attachments or copies, and you are prohibited from retaining, distributing, disclosing or using any information contained within. Please contact the sender and advise of the erroneous delivery by return e-mail or telephone. Thank you for your cooperation.
More information about the Histonet