[Histonet] microtome calibration
Kemlo.Rogerson <@t> waht.swest.nhs.uk
Wed Jun 3 09:31:27 CDT 2009
Have you tried altering the angle of the knife?
e-mail kemlorogerson <@t> nhs.net if not at work.
DD 01934 647057 or extension 3311 Mob 07749 754194;
Embrace uncertainty. Hard problems rarely have easy solutions. --Jonah
This e-mail is confidential and privileged. If you are not the intended
recipient please accept my apologies; please do not disclose, copy or
distribute information in this e-mail or take any action in reliance on
its contents: to do so is strictly prohibited and may be unlawful.
Please inform me that this message has gone astray before deleting it.
Thank you for your co-operation
From: histonet-bounces <@t> lists.utsouthwestern.edu
[mailto:histonet-bounces <@t> lists.utsouthwestern.edu] On Behalf Of Dianne
Sent: 02 June 2009 22:44
Subject: [Histonet] microtome calibration
Has anyone had problems with the calibration of an A/O Spencer sliding
microtome? I have used this one for years and now I am getting
'thick-thin' sections? I thought maybe the accumulation of oils in the
mechanics of it have slowed things down. Can WD40 be applied without
adverse effects? I certainly would not attempt opening up all the
gears!! Any suggestions would be appreciated or if a medical equipment
repairman is reading this - I really need this thing fixed.
Individuals who have received this information in error or are not
authorized to receive it must promptly return or dispose of the
information and notify the sender. Those individuals are hereby notified
that they are strictly prohibited from reviewing, forwarding, printing,
copying, distributing or using this information in any way.
Histonet mailing list
Histonet <@t> lists.utsouthwestern.edu
More information about the Histonet