[Histonet] thermometer for waterbaths
Marshall Terry Dr,
Terry.Marshall <@t> rothgen.nhs.uk
Tue Mar 20 11:22:09 CDT 2007
Odd that this subject came up now, because I had to come in at the
weekend, and went into the lab and looked at the waterbaths which were
still on. One had the thermostat (joke) set at 40 and the temperature
from a thermometer was 48. The other had the temperature set at 50, and
the temperature was 56.
And you lot have the cheek to make digs about the pathologists!!!
From: histonet-bounces <@t> lists.utsouthwestern.edu
[mailto:histonet-bounces <@t> lists.utsouthwestern.edu] On Behalf Of Vickroy,
Sent: 20 March 2007 13:52
To: histonet <@t> lists.utsouthwestern.edu
Subject: [Histonet] thermometer for waterbaths
I am wondering if there are any suggestions for waterbath (floatation
bath) thermometers. In the old days we had a submersible type but I am
wondering if there are some NIST traceable thermometers that are in use.
Digital or regular. We haven't decided whether to use one thermometer
for all of the baths or have a thermometer at every station. Ideas?
Technical Supervisor - Surgical and Autopsy Pathology
Memorial Medical Center
This message (including any attachments) contains confidential
information intended for a specific individual and purpose, and is
protected by law. If you are not the intended recipient, you should
delete this message. Any disclosure, copying, or distribution of this
message, or the taking of any action based on it, is strictly
Histonet mailing list
Histonet <@t> lists.utsouthwestern.edu
More information about the Histonet