[Histonet] Sakura Tissue Processor
j_gorenstein <@t> yahoo.com
Thu Jul 8 13:16:53 CDT 2004
Thanks LuAnn for the suggestion. However, we just had this instrument brought out of storage and we put in fresh filter (2 weeks old) and we change water and everything. It is useful to know that it should not be doing this, and that we should not need a hood. Any further suggestions are welcome! Thanks again!
LuAnn Anderson <ander093 <@t> tc.umn.edu> wrote:
Date: Thu, 08 Jul 2004 13:07:39 -0500
To: Juile Gorenstein
From: LuAnn Anderson
Subject: Re: [Histonet] Sakura Tissue Processor
Our VIP has a charcoal station in the very last reagent position. The
activated charcoal helps to absorb xylene fumes. If you are not using
charcoal in your VIP, or if it hasn't been changed in a while, I would
recommend filling that container with fresh activated charcoal pieces (not
powder). I think that I ordered it through Allegiance. It is Tissue Tech
Activated carbon #4663. If you need me to, I can look up where I ordered
and let you know for sure. Good luck.
At 10:43 AM 7/8/04 -0700, you wrote:
>I was wondering if anyone has had problems with xylene fumes using Sakura
>VIP Tissue processor. Ours is currently sitting on a bench in a room with
>a snorkle right above is and the smell is still really bad and is causing
>headaches for people who try to work in that room. Does anyone have any
>Novartis Institues for Biomedical Research
>Do you Yahoo!?
>New and Improved Yahoo! Mail - 100MB free storage!
>Histonet mailing list
>Histonet <@t> lists.utsouthwestern.edu
Do you Yahoo!?
Yahoo! Mail - 50x more storage than other providers!
More information about the Histonet