[Histonet] Superfrost plus slides
bills <@t> icpmr.wsahs.nsw.gov.au
Tue Dec 23 14:50:26 CST 2003
Merry Christmas to all from down under in Australia.
We to have had problems but only with limited numbers of slides. We were
told that there was "a" batch of Superfrost Slides which were suspect. I
found out that nearly all Superfrost Slides are manufactured in one or two
factories in Germany and packaged for the local distributors. The company
claimed they knew nothing about problems with any of their slides, however
there were two batches which gave problems one was # 3421272 and I believe
the other also ended in 72.
This happened about 3-4 months ago now and I have noticed that all the
batches have expiry dates on them.
>From correspondence on Histonet it appears that the problem or something
like it has been going on for several months.
But as is usual in Histotechnology everyone tries to solve the problem
in-house without discussing it with anyone else.
This is where I find Histonet invaluable.
Tissue Pathology, IMPAR
Westmead NSW 2145
Ph 02 9845 7774
From: histonet-bounces <@t> lists.utsouthwestern.edu
[mailto:histonet-bounces <@t> lists.utsouthwestern.edu]On Behalf Of
nick.kirk3 <@t> btopenworld.com
Sent: Wednesday, 24 December 2003 12:33 AM
To: M.Bromley <@t> dgri.scot.nhs.uk; histonet <@t> lists.utsouthwestern.edu
Subject: Re: [Histonet] Superfrost plus slides
We've have a similar problem on occasions with Surgipath slides as well. We
also have the DAKO Autostainer and use DAKO diluent and detection reagents.
> from: Mike Bromley <M.Bromley <@t> dgri.scot.nhs.uk>
> date: Tue, 23 Dec 2003 11:22:07
> to: scoticc <@t> bigfoot.com, dodson <@t> liv.ac.uk,
histonet <@t> lists.utsouthwestern.edu, suzy.howard <@t> dakocytomation.co.uk,
katrina.murphy <@t> dakocytomation.co.uk
> subject: Re: [Histonet] Superfrost plus slides
> Hi All
> We have had a problem with a batch of Menzel Superfrost Plus slides
> (3451272). They have been made in a way that makes them more hydrophobic
> than usual.
> The problem may not have been noticed by others because it only occurs
> certain circumstances, however the consequences can be extremely serious.
> The circumstances are:
> 1. Use of a "flatbed" stainer where the slides are held horizontally
> 2. Use of a bad batch of slides
> 3. Use of these slides for antibodies which do not require pressure
> cooking, this usually means enzyme treatment or no treatment (sometimes
> when heat treated in a 95-99* water bath as in the Herceptest) Pressure
> cooking the slides seems to overcome the hydrophobic nature of the "bad"
> Wash buffer usually spreads okay on the slides, but when the reagent is
> dripped onto the slide it fails to spread evenly. This is usually not a
> problem if the section is a single piece of tissue and it is hit by the
> drops however if the section is composed of small pieces then some of
> can be missed. Hence your controls may be fine, because they are often a
> single piece of a reasonable size, but you may miss a critical piece of
> I advise everyone who uses a flatbed horizontal stainer for sections not
> requiring pressure cook antigen retrieval to watch the addition of the
> dropped reagents. This problem is due to batch variation and may be slight
> or extreme depending on the batch.
> Other users have noted problems with Superfrost Plus slides before (see
> histonet archives) We use Superfrost Plus slides because they are better
> than home made APES slides especially for our slides that are pressure
> cooked in high pH solution.
> All our reagents are from Dako and the Autostainer protocol is followed.
> Best Wishes
> Mike Bromley
> Chief Biomedical Scientist
> Dumfries & Galloway Royal Infirmary
> Scotland, UK
> > This e-mail and any files transmitted with it are private and intended
> > solely for the use of the individual or entity to whom they are
> > If you have received this e-mail in error please return it to the
> > it
> > came from telling them it is not for you and then delete it from your
> > system.
Histonet mailing list
Histonet <@t> lists.utsouthwestern.edu
This electronic message and any attachments may be confidential. If you
are not the intended recipient of this message would you please delete the
message and any attachments and advise the sender. Western Sydney
Area Health Services (WSAHS) uses virus scanning software but excludes
any liability for viruses contained in any email or attachment.
This email may contain privileged and confidential information intended
only for the use of the addressees named above. If you are not the
intended recipient of this email, you are hereby notified that any use,
dissemination, distribution, or reproduction of this email is prohibited. If
you have received this email in error, please notify WSAHS
Any views expressed in this email are those of the individual sender
except where the sender expressly and with authority states them
to be the views of WSAHS.
More information about the Histonet