saby_joseph_a at yahoo.com
Wed Oct 14 18:44:35 CDT 2015
We had this happen when the individual changing the processor reversed order
of the alcohols, placing the newest first and the oldest last. Hope your answer
is as simple, even if it is embarrassing.
From: "Abbott, Tanya via Histonet" <histonet at lists.utsouthwestern.edu>
To: "histonet at lists.utsouthwestern.edu" <histonet at lists.utsouthwestern.edu>
Sent: Wednesday, October 14, 2015 4:35 PM
Subject: [Histonet] VIP5
Help!!! We changed our processor on Monday, all reagents. Run Monday into Tuesday a.m. was fine. Tuesday nights run had some blocks in it that weren't fully processed; chamber looked like it had droplets of residual water (or maybe alcohol?) in it. Dumped and refilled everything, ran a few test blocks, same scenario.
Tanya G. Abbott
Penn State Health St. Joseph
Histonet mailing list
Histonet at lists.utsouthwestern.edu
More information about the Histonet