[Histonet] Formalin smell in the last paraffin station in vip
6 tissue tek
j.rowaihi <@t> alborglaboratories.com
Tue Nov 25 02:18:44 CST 2014
If the paraffin are overheated it gives abnormal odor.
Recheck the paraffin bath temperature and the paraffin melting point then make sure to adjust the bath temperature at 2 degrees above the melting point.
Please update me.
Jamal M. Al Rowaihi Anatomic Pathology Supervisor | Al Borg Medical Laboratories | Mobile +966 503629832| j.rowaihi <@t> alborglaboratories.com
Palestine St, Al Rajhi Building, P.O. Box 52817, Jeddah 21573, KSA | Phone: +966 12 670 0099 | Fax: +966 12 676 4984 | www.alborglaboratories.com
From: histonet-bounces <@t> lists.utsouthwestern.edu [mailto:histonet-bounces <@t> lists.utsouthwestern.edu] On Behalf Of Arun Jyothi S.P
Sent: Tuesday, November 25, 2014 9:53 AM
To: histonet <@t> lists.utsouthwestern.edu
Subject: [Histonet] Formalin smell in the last paraffin station in vip 6 tissue tek
After processing in vip 6 the last paraffin has a strong odour of formalin
Have anybody experienced the same
Any ideas why it is happening.
Histonet mailing list
Histonet <@t> lists.utsouthwestern.edu
More information about the Histonet